Hanna HI 723 Checker High Range Chromium VI Photometer HI723
Item Description: Brand New Hanna Instruments HI 723 Checker®HC Chromium VI HIGH RANGE Photometer 0-999ppb
FINALLY! A SIMPLE HIGH RANGE CHROMIUM VI TEST!
Checker®HC Handheld Colorimeter - Chromium VI High Range
|Range||0 to 999 ppb|
|Accuracy @ 25°C||±5% of reading ±5 ppb|
|Light Source||LED @ 525 nm|
|Light Detector||silicon photocell|
|Environment||0 to 50°C (32 to 122°F); RH max 95% non-condensing|
|Battery Type||(1) 1.5V AAA|
|Auto-off||after ten minutes of non-use|
|Dimensions||81.5 x 61 x 37.5 mm (3.2 x 2.4 x 1.5”)|
|Weight||64 g (2.25 oz.)|
|Method||adaptaion of the ATSM, Manual of Water and Enviornmental Technology, D 1687-92, Diphenylcarbohydrazide method|
- Easier to use and more accurate than chemical test kits
Accuracy ±5 ppb ±5% of reading
1 ppb (µg/L) resolution
Large, easy to read digits
Auto shut off
- Dedicated to a single parameter
Designed to work with HANNA’s powder reagents
Uses 10 mL glass cuvettes
- Small size, big convenience
Weighing a mere 64 g (2.25 oz.), the Checker®HC easily fits
into the palm of your hand or pocket
Use for quick and accurate on the spot analysis
Single button operation: zero and measure
Operated by a single AAA battery
- Ideal for Water Quality
There are two natural forms of ionic chromium, the hexavalent ion, Cr (VI) and the trivalent Cr (III). Cr (III) is much less toxic than Cr (VI) and seldom found in potable waters. Cr (VI), however, is toxic to humans and is found in water. Even though the toxic effects from Cr (VI) in drinking water are not well documented, it is a suspected carcinogen. The maximum permissible level or Cr (VI) allowed to be released into the waterways is 50 ppb. Its level in drinking water is normally much lower and a lever higher than 3 ppb is suggestive of industrial pollution.
By monitoring this parameter with the HANNA HI 723 Checker®HC, serious health and safety risks can be avoided.
The HI 723 Checker®HC is extremely simple to use. First, zero the instrument with your water sample. Next, add the reagent, shake vigurosly for 10 seconds. Last, place the vial into the Checker®HC, press the operational button for about 3 seconds and the display will show the countdown prior to the measurement or, alternatively, wait for 6 minutes and press operational button. When the timer ends the meter will perform the reading. The Checker®HC displays concentration in ppb of chromium VI. It’s that easy.
HI 723 Checker®HC is supplied with sample cuvettes with caps (2 ea.), power reagents for chromium VI HR (6), battery and instructions.
We also carry the other Hanna Checkers -
Look to NI Supply for more details later...
National Industrial Supply
30777 Rancho California #891420
Temecula, CA 92589, USA
We Also Recommend