Hanna Instruments HI70007P pH Meter Buffer Solution Pouches 7.01, 20ml Pouch
Item Description: Hanna Instruments HI70007P pH Buffer Solution in a 20ml Pouch, 7.01pH
Use these products to keep your pH Meters running at optimum levels!
This BIN is for only ONE Packet at a time, but you can buy as many as you need!
7.01pH - HI70007P
IF YOU NEED OTHER pH's, PLEASE LOOK IN OUR eBay STORE FOR THEM, OR CONTACT US!
IF YOU NEED (10) POUCHES, ORDER (10) FROM THIS LISTING!
To ensure optimum accuracy time after time, calibrate your instruments using the pH Buffer Solutions. Ideal for 1-Point, 2-Point or 3-Point Calibrations! Don't buy Kool-aid type pouches that rely on you having access to deionized water or reverse osmosis water and lab equipment to mix the solution. Buy smart, buy a premixed product to save you time and peace of mind!
* 20 ml Foil Pouches
* Temperature Charts are included on each Pouch of solution.
* Extended Expiration Dates! Expiration dates on every pouch, usually with more than 24 months left.
* pH 7.000 is the most commonly used Buffer Solution.
Proper electrode storage maximizes electrode performance and extends electrode life. The electrodes should not be stored in distilled water. The filling solution will be diluted and the electrode response will be slow. If using a refillable pH electrode, the fill hole should be covered overnight and during storage. The fill hole cover must be removed during calibration and measurement to ensure a uniform flow of electrode filling solution. The best way to store the electrode is to read the included manual.
Maintenance and cleaning: The electrode should be inspected weekly for scratches, cracks, or salt crystalization. Follow instructions and do not clean excessively.
Calibration: To verify electrode slope and proper function, calibration must be performed each day before any samples are measured and recalibrated every couple of hours to ensure that correct slope is used to calculate the sample pH. If the user is unsure of the sample range, a pH test strip can quickly determine the approximate sample pH. TIP: Regularly, the electrode should be calibrated with 2 or 3 buffers or standards that bracket the expected sample range.
Rinsing between measurements: Rinsing prevents contamination by carryover on the electrodes. Tip: The electrode should be rinsed with deionized water or an aliquot of the next buffer or sample. Caution: the pH electrode glass bulb should not be wiped.
Temperature: To account for pH slope, buffer and sample changes, a separate or integrated automatic temperature comp. probe or digital LogR should be used. pH is readily temperature correctable.
Consider a storage solution for your probe when not in use! We carry that in our eBay Store at www.nisupply.com.
National Industrial Supply
30777 Rancho California #891420
Temecula, CA 92589, USA